CB9 CARELINE® The CB9 CARELINE® has been developed for use together with LA31/LA31R, LA34/LA34R* in the Care & Rehab industry. CB9 and LA31 can be fully integrated which saves mounting and wiring or be installed separately. Exchangeable mains cables, Electronic Overload Protection (EOP), EAS, earth connection (Class 1) and exchangeable mains fuse makes CB9 a good choice for the simple hospital and care beds. * The combination of LA34R/CB9 ”PX” is only available as a special article. Features and options: • Connection of max. 4 actuators, type LA31/LA31R and LA34/LA34R, via 4-pole DIN sockets. • Extremely compact design. • Mains voltage: 100/120/230V AC 50-60 Hz • Output voltage 24V DC • Protection class: IPX4 or IPX6 (not for wash tunnels) • Colour: Grey or black • DIN-socket for handsets: - Ax types : HB40, HB60, HB70, DP or FS - Px types : HB4XE, HB5XE, HB60M (PM/PN versions), HB7XE/W, DPXE or DP1M • 3.2 m exchangeable straight mains cable • EOP = Electronic Overload Protection. - AJ and AL types CH1 + 2 (3.4A) and CH3 + 4 (7A) common current cut off. - AS type via individual current cut off. • Px type via pulse measurement. • Locking mechanism for DIN plugs and mains cable. • Electronic protection of the relays for extended lifetime, EAS (simple EAS); available on AC, AJ, AK, AF, AL, AM and Px (not PM and PN) types. • Double power control (FET + relay) monitoring on all AC, AJ, AK, AF, AL & AM types • Class 1 or 2 protection LINAK.COM/MEDLINE-CARELINE • Power supply indication via an LED. Not available with mains cut-off. • Exchangeable mains fuse from outside • External battery back up and internal charger (only AC, AJ, AK, AF, AL and AM types; use notation A for ordering) • External battery back-up (no internal charger) (only AC, AJ, AK, AF, AL and AM types; use notation B for ordering) • External battery back-up alarm (only AC, AJ, AK, AF, AF and AM). • Short cut protection: only AC, AJ, AK, AF, AL and AM (15A) Only available in Px types: • Choice of 8 different standard software packages. All standard software packages have either impulse memory or memory. To use memory in general a handset with memory has to be chosen. • Energy saving function (the control box restores to power down when not operated) • Compact high power toroidal transformer with low power consumption and low electromagnetic emission • All actuators, which are used together with Px types, must be equipped with reed switch and connected with a DIN-plug.
Download PDF file